When your system indicates a flow alarm, this means that the air flow of the air drawn in is not correct during measurement. During measurement and calibration, the air flow should be approximately 1 liter per minute. If this is not the case, the system will indicate a flow alarm. The air flow can be read on the flow meter on the front of your control cabinet.
Watch the video to see step-by-step how to solve the alarm.
If you still have questions after watching the video or if you cannot solve the problem yourself using these instructions, please contact your installer.
We are constantly trying to improve the videos. If anything is unclear, we kindly request you to mention this in the feedback field on the right.
When the system indicates a flow alarm on several rooms at the same time or when there is no suction from the control cabinet, this means in most cases that the problem is in the cabinet and you can check your system for the following points:
1) Is the moisture separator properly closed?
2) Check if the microfilter or moisture filter is dirty and may need to be replaced.
3) Check if no hoses are kinked in the measuring circuit of the control cabinet.
4) Check the operation of the measuring pump. It may not be powered.
Do you have a flow alarm on a single storage room or do you have suction from your control cabinet? Then watch the video below.